INTRODUCTION FOR USE (ABUTMENT)
1. Purpose of product
This product is connected to fixture and for support
prosthetic restoration of lost tooth.
It is a disposable sterilized medical
Cautions of use
only skilful dentist, use only one time.
the term of validity and the package condition.
dentist and dental technician make a surgical plan by discussing to accurate examination
of patient’s condition.
a plaster cast after taking impression model by occlusion with upper and lower
on cover screw or healing abutment in mouth of a patient until osseointegration.
relationship (insufficient bone quantity), problematic occlusion or functional
relationship, pathologic condition of alveolar bone, radiation therapy on
alveolar bone, xerostomia, pathologic change of oral mucosa (Leukoplakia,
lichen planus, stomatitis), Macroglossia, Unrestored tooth, poor oral hygiene.
that is transiently limited
Acute inflammatory disease and
infection, pregnancy, temporary medication of specific medicine
(Anticoaggulant, immunodepressive), physical or mental stress.
Patients with poor cooperation, alcohol
and drug abuse, neurotics, patients with mental illness.
Current medication (corticosteroid, long
term antibiotic treatment), metabolic disorder (juvenile diabetes, blood sugar
level – higher than 300), Hematologic disorder (RBC, WBC, Coaggulation system
disorder), metabolic disorder of bone (Osteomalacia , osteitis deformans,
osteogenesis imperfecta, menopausal osteoporosis), Collagen disorder
(Dermatosclerosis, rheumatic arthritis), implant as focus of potential
bacterial growth (Artificial heartvalve, bacterial endocarditis).
prepare before use
period for osseointegration is 3-6 months after inserted fixture .
cover screw and connect healing abutment after healing period for
for healing period of soft tissue.
soft tissue is healed (In one stage surgery, it is possible right after
removing healing abutment), take impression with impression material.
and size of abutment is decided by X-ray taking impression and relationship of
this product is not sterile, it should be sterilized before installation to
mouth. Autoclave in dental clinic is used and it should be longer than 30
minutes in 121°C
of abutment in mouth.
final impression in installation abutment.
selected abutment is cemented type, wax up the abutment by plaster cast, so
restoration take on fixture or abutment in mouth, check occlusion with upper
and lower surface.
torque is 30-35N/cm2 when connecting abutment with fixture.
torque is 8-10N/cm2 when connecting healing abutment and cover
handling after use
Use only one time. Do not reuse.
12°C ~ 28°C and 0 ~ 80% in cold, shade and well ventilation
in room of chemicals storage, gas issues and fearing infection.